All rights reserved. Printed in the Netherlands. All product names are trademarks of their respective companies. Fluke // ScopeMeter. Users Manual. Items are supplied with manuals, accessories and typically a full no-quibble 2 . 2 Fluke Corporation ScopeMeter C/B Series, Series and the NEW. Path: /fluke/B/. This is the These manuals are available for download and free of charge. Fluke B/B-C/B-C ScopeMeter Service Manual.
|Published (Last):||13 July 2014|
|PDF File Size:||6.74 Mb|
|ePub File Size:||14.85 Mb|
|Price:||Free* [*Free Regsitration Required]|
It performs an FFT to transform the amplitude waveform from the sscopemeter domain into the frequency domain. Triggering on Waveforms Using Trigger Delay or Pre-trigger Using Trigger Delay or Pre-trigger You can begin to display the waveform some time before or after the trigger point has been detected. Turn the test tool off.
FLUKE B USER MANUAL Pdf Download.
When fully charged, the batteries provide 4 hours of use. Start on trigger to start recording; Meter mode has the advantage of scopemwter measurement resolution. As soon To reduce jitter on the screen when triggering on noisy as the test tool receives a trigger, the waveform is waveforms, you can use a noise rejection filter.
You can display a fixed reference waveform with the – the last used reference waveform actual waveform for comparison. To activate manual ranging, do the following during any Meter measurement: On or Off to You can use Persistence to observe dynamic signals.
Page 7 Contents continued Triggering on Video Signals The set includes the following items not available separately: Read First Warning To avoid electrical shock or fire: Accept the current measurement. The Spectrum function shows the spectral content of the input A or input B waveform. Using Average for Smoothing Waveforms Exit the menu. Enable the arrow keys to adjust the pulse width.
To recall the setup configuration amnual memory 1, do the following: The introduction does not cover all of the capabilities of the scope functions but gives basic examples to show how to use the menus and perform basic operations. See next This chapter provides an introduction to the trigger section. Scopemwter product names are trademarks of their respective companies.
After a reset, the test tool is dc-coupled so that ac and dc voltages appear on the screen.
From scopemetsr position the following: For complete service, not use abrasives, solvents, or alcohol. The display also shows the actual waveform. Comments to this Manuals Your Name. You can use the average functions to suppress random or Display the key labels. Use caution when using 1: Don’t have an account? Subsequently follow steps 1 through 4 to open the scope menu and to choose an item. The test tool offers a wide range of automatic scope To choose also a Peak-Peak measurement for Input B as measurements.
The test tool automatically calibrates itself to the probe. Since the commons are not directly connected, the chance of causing short circuit when Isolated Inputs measuring multiple signals is greatly reduced. Tips Using the Tilt Stand Using the Scopemetsr Stand Resetting the Test Tool The test tool is equipped with a tilt stand, allowing viewing If you want to reset the test tool to the factory settings, do from an angle while placed on a table.
You can select the language in following: DC to 20 kHz Note To maximally benefit from having independently Figure 6. This averages Capture the outcomes of 64 acquisitions.
Chapter 3 Using The Recorder Functions About this Chapter This chapter provides a step-by-step introduction to the recorder functions of the test tool. A marker is To measure rise time, do the following: The following tables list the user-replaceable parts for the Display the key labels.
Fluke 192B User Manual
This example demonstrates how to set up the test tool for Consult the manual that came with your printer to find out printing on a postscript printer with a baud rate: You can set the power shutdown time to 5 minutes as following: The introduction gives examples to show how to use the menus and perform basic operations.
Read First Safety Information: From scope mode, display the cursor key labels. Perform an auto set. This option gives the best printing results.